Call us 24/7 at 800-860-6272
or email us
wind turbine kit,
blood type test,
What is a
Compound Microscope? This is the most common type of microscope, found
in any school or microbiology lab. It's used to greatly
magnify tiny specimens like amoebas or thin sections of specimens like leaves or
muscle tissue mounted on microscope slides. A compound microscope is ideal for
observing cells and cell structure. Almost every student will use a 400x
compound microscope to study biology.
& Dissecting Microscopes? These have lower magnifications of 15 to 45x,
and are designed for viewing whole, three-dimensional objects like rocks,
insects, flowers, and dissection specimens. They are important for advanced
nature studies and ideal for children since an object can be viewed without
preparing a slide - simply stick it under the microscope.
Terms and Parts of a Compound Microscope
(See these parts on the microscope diagram shown below the glossary.)
explains the main parts of your compound microscope and the function of each:
Condenser: The light rays from the microscope's illuminator (light) are condensed and focused
through this lens in the center of the microscope stage, providing better image resolution.
Most 1000x scopes have a built-in, movable 1.25 Abbe condenser underneath the stage.
- Disc diaphragm or Iris diaphragm: The disc
diaphragm is used on most microscopes without 1000x. It's a
rotating disc fixed under the stage, designed to provide the best resolution for
the objective lens. Turn the disc to the smaller holes when using low magnification
(40x to 100x) and to the larger holes for high magnification (400x). Most
1000x models have a fully-adjustable
iris diaphragm with an
easy, sliding control lever rather than a rotating disc. This type of diaphragm
allows you greater control of contrast and helps achieve high image resolution.
- Eyepiece: The part of the microscope
that you look through. Eyepieces in most compound microscopes have a lens with
a 10x magnification level (although you can upgrade with a 15x eyepiece). The
National Optical 131
microscope has a pointer built in, which appears as a black line across half of the field of view. To
move the pointer, just turn the top of the eyepiece.
- Field of view: This is the area that's visible
when you look through the eyepiece. The greater the magnification used, the smaller the
field of view (i.e., FOV at 40x is much greater than at 1000x. This is why you
should locate a specimen at lowest power and then gradually narrow your focus
as you are increasing magnification).
Filter holder and filters: Some microscopes (like our
Home 1000x) come with a
swing-out filter holder at the bottom of the condenser. Blue and yellow filters
can be used to enhance the image in some situations.
- Focusing: Turning the focus knob or knobs moves the microscope stage
up and down to bring the slide sharply into view. Most microscopes
have both fine and coarse focusing, either as two separate knobs or else
coaxial (one on top of the other for easy reach).
- Head: The standard head type is
monocular - just one
eyepiece. Some microscopes will have a dual head so a teacher
and student can view a slide at the same time (it's also convenient for video/digital photography
with a digital microscope camera). The most
deluxe models have
binocular heads for extra-comfortable viewing.
Illumination: There are several types of electric lighting for
Tungsten is the least expensive illumination. It is hotter and
less bright than the other kinds.
Fluorescent illumination provides cooler and brighter light than
tungsten. This is beneficial when viewing slides for long periods of time or
observing live specimens, such as protozoa.
LED is a bright cool source of light with a bulb that is much longer lasting than a fluorescent bulb. Microscopes with LED lighting typically have a variable control allowing you to change the intensity of light.
Halogen provides the very brightest illumination.
The best microscopes have halogen lighting
stereo microscopes with top lighting also use halogen lighting. The intensity of halogen light can be adjusted with a variable control.
- Lenses: Standard
achromatic lenses help prevent color distortion. Semi-plan or plan
lenses improve image quality through superior clarity and flatness. Super high
contrast lenses offer extra
image contrast so that the image appears much sharper.
Magnification: Multiply the
magnification of the eyepiece by the magnification the objective lens for the
total magnification at that power. If you have a 10x eyepiece and are looking
through the 4x objective, the specimen you see is magnified 40x its actual
size. Magnification of 400x or 1000x is necessary for
studying cells and cell structure.
stage: This fits onto the stage (in some microscopes it is
built in) and provides the best slide
control, which makes viewing easier. Clip the slide into place and use the X
and Y-axis knobs on the stage to move the slide back and forth under the objective lens.
- Objective lens: This is
second lens of the microscope after the eyepiece. Compound microscopes have a "nosepiece" with a
rotating objective turret, which allows you to change the magnification level
for different specimens. The standard objectives are 4x, 10x, and 40x for total
magnification of 40x, 100x, and 400x. DIN is an
international standard of lens quality.
objective: Some microscopes have a fourth objective, a 100x oil-immersion lens for
1000x magnification. This allows you to see greater cell detail and cell
structure. To use this lens, you need to put 1-2 drops of immersion oil on
the slide coverslip. Immersion
oil prevents light distortion, so that you can see a sharp image even
with the lens pressed right up against the slide.
This means that if you centered your slide
while using one objective lens, it should still be centered even when you switch to
that once you've focused on an object using one objective, the microscope
will still be coarsely focused when you switch to a different objective. All
you'll have to do is adjust the fine focus a bit.
- Stage: This is the platform
that holds the slide up beneath the objective lens. The stage
clips hold the slide in place. (In some microscopes, these
are replaced by a more convenient mechanical stage).
Stage stop: This is the small bar and screw between the stage and
the arm of the microscope. It prevents the stage from coming too far up, so
that the stage can't grind against the objective lens. This is also known as a
"safety rack stop," and is pre-adjusted by the manufacturer although it can
be adjusted by you if need be.
See a glossary of terms for stereo microscopes below.
Diagram of Compound Microscope Parts
Terms and Parts of Stereo Microscopes
The following list explains the
basic stereo and dissecting microscope parts and functions of each part.
head: Most low-power microscopes have
what is called a stereo head, which has two eyepieces - looking through them is
something like looking through a pair of binoculars. Some stereo microscopes
only have a monocular (single) eyepiece.
- Eyepiece: This is the part of the microscope that
you look through. Eyepieces in stereo microscopes typically have a lens with a 10x magnification level.
(You can increase power with a 20x eyepiece upgrade, though.) Both
eyepieces have the same magnification on stereo head microscopes.
- Diopter: In binocular eyepieces, this compensates for focusing
differences between your eyes so you see the image clearly through both eyepieces.
Objective lens: The objective lens together with the lens of the eyepiece make up the
microscope's magnification. Some stereo microscopes have a rotating objective
turret (such as the one pictured below), which allows you to change the
magnification level for different specimens; others have a set objective. Stereo
microscopes actually have two separate objectives so that each eye is
looking through an eyepiece lens as well as an objective lens.
Focus knob: Moves the head of the microscope
up and down to bring the object sharply into view. Most stereo microscopes have
only one focus knob.
Lighting: Top lighting shines down and
reflects off opaque or solid specimens; bottom lighting shines up through
transparent objects. The microscope pictured below has top and bottom lighting,
though not all microscopes will have both.
Rack and pinion focusing: Most stereo and dissecting microscopes have standard "rack and pinion" focusing. Turn
knob to slide the head of the microscope up and down (closer or farther from the
Stage clips: These hold microscope slides (which can be viewed under a
stereo microscope as long as you don't need to see fine cell details) or
other thin objects in place on the stage.
Stage plate: Where the specimen is placed for
viewing; located directly under the objective lens. Some stereo microscopes have
reversible black and white stage plates to provide appropriate contrast with
the object being viewed. (Black for light-colored or see-through specimens and
white for darker or opaque specimens.)
Diagram of Stereo and Dissecting Microscope Parts
The diagram below shows where each
part is located on a typical stereo microscope:
Click here to
see a comparison chart of compound microscopes available from Home Science
Tools, or click here
for a comparison chart of stereo microscopes.
Return to top of page